BIOHAZARD SAFETY MEDICAL PAPER CARDBOARD SHARPS CONTAINER BOX IN NIGERIA BY SCANTRIK MEDICAL SUPPLIES
Sharps safety boxes, constructed of cardboard or other materials, flat-packed, one piece with aperture and handle.
Intended use: safely and efficiently contain, transport and store used sharps injection devices until final destruction, safe disposal or recycling.
Minimum dimensions: The minimum height from the bottom of the container to the fill line is not less than 230mm.
Functionality: The safety box must safely contain contaminated sharps at the point of use; during temporary storage and during handling and transport to the point of treatment and final disposal.
Nominal Capacity: Boxes must accept no less than 20 units of 0.5ml AD syringes per nominal liter. Total storage capacity not less than 100 units/ 5L box. No syringe must protrude from the container or above the fill line and the box must be capable of being correctly and permanently closed without any risk of needle-stick injury
Maximum capacity: allowed to exceed the nominal capacity of 20 syringes per nominal liter provided no syringe must protrude from the container or above the fill line and the box must be capable of being correctly and permanently closed without any risk of needle-stick injury
Sharps aperture: Boxes fitted with a sharps aperture of 38 mm diameter, or 38mm width and breadth and placed at least 50 mm above the maximum recommended fill line marked on the exterior of the box. It is possible to close and seal the aperture at any time between empty and full to maximum capacity.
Handles: positioned above the fill line, does not obstruct access to the sharps aperture, sufficiently robust. Filled safety box is safely carried with one hand
During transport to the disposal site.
Color: color of unbleached sulphate board, or non-chlorine bleached white, or yellow.
Fill line: clearly marked on all vertical faces of the box, in black or red.
Shipping and storage volume before use:
Boxes must be supplied flat-packed or nested to minimize shipping and storage
Marking instructions on each safety box:
Bio-hazard marking: marked with the international bio-hazard warning not less
than 50mm diameter, printed in black or red on each of the front and back faces of
Pictorial instructions: without writing printed on two sides of the container showing:
• How to assemble the box.
• How to use the box as a container for contaminated sharps;
• Syringe disposal direction (needle down).
• How to close the sharps aperture when the box is full.
Product and packaging: Product listed under WHO Performance Specification (PQS) Category E010 waste management equipment for immunization. Verified in accordance with PQS Verification Protocol E10/SB01-VP
Quality management system (QMS): Products and manufacturer must be certified and comply with ISO 9001: 2008 requirements (or equivalent).
We are your One Stop Hub for Medical Equipment | Hospital Supplies | Lab Consumables | Surgical Instruments | Dental Care | Clinical Devices | PPE Kits | Disinfectants | Consumables & Disposables |Lab Diagnostics & Instruments | Medical Device & Equipment |Medical Implants | Hospital Establishment | Physiotherapy & Rehabilitation-Medical Aids | Mobility Aid | Medical Furniture | Gynaecology & Infant Care | Hospital Holloware | Lab Reagents | Radiology Equipment (Ultrasound, X-ray Machine, CT, MRI, ECG)
WE CAN SUPPLY AND INSTALL ANYWHERE IN NIGERIA
OFFICES IN LAGOS, ABUJA, PORT HARCOURT AND BENIN CITY
Please CALL: 0907-643-3020, 0904-936-7703
Email: info (AT) scantrikmedicalsupplies.com
- Service Type : Wellness & Beauty